- freely available
Color Parameters of the Chromascop Shade Guide
AbstractThe purposes of this study are: (1) determine the color of the twenty shades in the Ivoclar’s Chromascop shade guide, (2) determine the color representation of the shade guide described as coverage error (CE), and (3) compare this shade guide with the Vita Classical and Bioform shade guides. The spectral data was collected using Beckman model DU reflectance spectrophotometer equipped with an integrating sphere. Commission International de l’Eclairage (CIE) chromaticity coordinates were calculated using CIE illuminant C and 1931 observer data, then converted to CIE L*a*b* and Munsell notation. Each shade was spectrophotometrically compared to the published colors of 335 human teeth. The minimum CIE L*a*b* color difference was calculated for each tooth and the average of these color differences was defined as the CE. The measured colors of the Chromascop guide had a CIE L* range of 79.67 to 65.61, an a* range of -0.71 to 3.85, and a b* range of 14.58 to 27.69. The average CE of the Chromascop shade guide was 3.38. The Chromascop shade guide has similar colors and a CE compared with the Bioform and Vita Classical shade guides, but with some shades of higher red and yellow components.
Share & Cite This Article
O'Brien, W.J.; Oh, W.-S.; Piché, P.W. Color Parameters of the Chromascop Shade Guide. Dent. J. 2013, 1, 3-11.View more citation formats
O'Brien WJ, Oh W-S, Piché PW. Color Parameters of the Chromascop Shade Guide. Dentistry Journal. 2013; 1(1):3-11.Chicago/Turabian Style
O'Brien, William J.; Oh, Won-Suk; Piché, Peter W. 2013. "Color Parameters of the Chromascop Shade Guide." Dent. J. 1, no. 1: 3-11.
Notes: Multiple requests from the same IP address are counted as one view.